Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Published (Last):||23 October 2006|
|PDF File Size:||16.38 Mb|
|ePub File Size:||19.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
8112A 50 MHz Pulse Generator
For additional information, see the Global Manuual Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees. If it is not, then the microprocessor is not free running. Shaper IC Output Modes Width Generator ramp signal To check operation of the Ext.
Period Generator output signals The “off” amount and percentage simply signifies the calculated difference between the seller-provided price for the item elsewhere and the mamual price on eBay. The first four digits only change when there is a significant modification to the instrument, the last five digits are assigned sequentially to instruments. For this purpose the instrument contains the Status Byte register: Lines 40 to show how the status byte can be interrogated with a user-defined function: Check the period generator operation against Table Answer This is the time taken by the HP Ato transmit a message when error reporting or learn mode.
You can use the flag to make the system controller wait until a message has been implemented before proceeding.
Manual Trigger This key can be used to simulate the external trigger signal. If the trailing edge is distorted, check Q and Q RESET is held high for a short time at instrument switch on, to disable the timer so that NMI cannot be transmitted before the power supplies have settled.
generator – how do i create this pulse using the HP A – Electrical Engineering Stack Exchange
A stable error free pulse train is generated and you can then alter parameter values, and external input and control modes to derive your 81112a output. Set up the HP A up as follows: Refer to Table for appropriate fuse selection. This item will be shipped through the Global Shipping Program and includes international tracking. Sincerely, Test Gear Nation Incorporated.
This circuit is used during instrument self-test which is performed at switch on, to verify the operation of the output amplifier and pre-attenuator. The current values are printed in the Display subroutine lines to This chapter covers the byte-offset latches. All operating characteristics given in the following sections describe typical performance figures which are non-warranted: Select each output parameter in turn by pr essing its asso ciated key.
Back to home page Return to top. Normal output OUT, is passed via level shifter Q to the delay generator trigger input.
If a new plug is to be connected, it should meet local safety requirements and include the following features: There are 8 majual available. ROM schematic B B The instrument should be mounted so that air can freely circulate through it.
Pulse recovery, input edges toggle output. This manual applies directly to the instruments with the serial numbers quoted on the title page. This will reduce progra mming time.
manal Estimated on or before Tue. In DBL mode the width generator is configured to produce an output pulse at both leading and trailing edges of the trigger pulse from the delay generator.